Read or Download Agilent 16517A, 16518A Synchronous State Logic Analyzer (service) PDF
Similar logic books
This vintage textual content has brought tens of millions of scholars to sound reasoning utilizing a wealth of present, proper, and stimulating examples all prepare and defined in a witty and invigorating writing kind. lengthy the alternative of teachers who are looking to "keep scholars interested", common sense AND modern RHETORIC: using cause IN daily life, 11E combines examples from tv, newspapers, magazines, ads, and our nation's political discussion.
Explores Schelling's Essay on Human Freedom, targeting the subjects of freedom, evil, and love, and the connection among his principles and people of Plato and Kant.
- Learning Language in Logic
- Logic Programming and Nonmonotonic Reasoning: 12th International Conference, LPNMR 2013, Corunna, Spain, September 15-19, 2013. Proceedings
- Generalized Recursion Theory II: Proceedings of the 1977 Oslo Symposium: Symposium Proceedings
- Secrets of Sankhya
Extra info for Agilent 16517A, 16518A Synchronous State Logic Analyzer (service)
Press the Peak-to-Peak Voltage key. Record the Delta-V value at the bottom of the oscilloscope display in the voltage swing fields in the performance test record. Test the next pod Beginning with the procedure on page 3-19, "Set up the logic analyzer," continue the test for the rest of the pods, substituting the pod to be tested wherever the test mentions master card pod 1. 3–19 Testing Performance To test the minimum clock period To test the minimum clock period The Minimum Clock Period test utilizes the Agilent Technologies 8131A pulse generator in 1 GHz transducer mode to supply the 16517A master board clock channel with a 1 GHz signal.
3–10 Testing Performance To test the threshold accuracy Set up the logic analyzer 1 Select the Acquisition mode field. In the pop-up menu, select State Acquisition Mode: Full Channel. 2 Touch the field showing the channel assignments for master card pod 1. Activate all channels by selecting Clear (if all channels are not activated with asterisks (*) showing in the channel position). 3 Touch the field showing the channel assignments for master card pod 2. Deactivate all channels by selecting Clear.
Deactivate all channels by selecting Clear. Press Done. d Touch the field below master card pod 1, then select ECL in the pop-up menu. 2 Touch Format, then select Trigger. In the Trigger menu, touch Modify Trigger, then select Clear Trigger, then select Clear All. 3 Select Trigger, then touch Waveform. Select the sec/div field. At the numeric pop-up menu, type in 50 ns. 3–17 Connect the logic analyzer 1 Using the 6 x 2 test connectors and BNC adapters, connect the data channels of master card pod 1 to the pulse generator output.